There are currently no product reviews.
let's say first that i do not need to have a credit for my opinion, i am a retired sparkie and i voluteerd to fix an electronic device for a local "Youthgroup",as no diagram was present i checked the "net" and gambled on this site and paying some fee via PayPall, i was gladly surprised that the manual arrived as was stated, GOOD SHOW, and best wishes, John
I had been looking for a Manual for my CS2150 for quite a while -- in fact I had just about given up. I saw this site and decided to download the Manual. When I Received it by Email I was really pleased with what I got, with the result that My Kenwood 'Scope is now 100% repaired and working well. As an AV Serviceman, you need a good 'scope, and thanks to this site, and the Service Manual, I have been able to repair it. The Manual was a copy of the Factory Original and the copy was very clear, especially in the area of the Circuit Schematics, where You really need to be sure of what You are looking at.
I recently purchased a manual for a Samsung DLP tv to help with a trouble shooting problem I was having. Every tv repair shop wanted close to $400.00 for the fix, but after I found Owner-Manuals.com I hit pay dirt. The manual they had for me to purchase and download was a complete service manual for the exact tv I needed. It was complete with wiring diagrams, schematics, and even part numbers. If your the fix it yourself type I highly recommend trying to find any manual here before paying someone else to fix whatever problem your having.
Once again, excellent price and manual delivered in a timely manner and as advertised!
Outstanding quality manual. This is the exact documentation I needed to service my AKAI GX-210D. This is a PERFECT COPY of the service manual for my machine. Outstanding service. Thank-you!
6.1.3 MD Mechanism Assy Adjustment and Check
1 Temperature Check (Please perform this check soon after the power has been switched on.)
Note: When IC101 (RF-AMP) or IC104 (mechanical controller) has been exchanged or when a correction changing VD+3 has been made, be sure to correctly perform all steps up to step 7.
Step No. 1 Operation Keys and Operation Method Short circuit MECH TEST MODE during the power ON, or press the ESC key. (Service remote control unit)(Fig.1) Press the REC key. When is 80 or more at this time, the adjustment is OK (step 4 and following can be omitted). is 80 (not passed) Measure the ambient temperature T (Â°C) andcheck hat the measuring value ( ) is within the standard in regard to the microcomputer set value ( ). Note: When the indication of the measuring value ( ) is not stable, and the ambient temperature T (Â°C) is 25 Â°C or less, set to the lower value. When the ambient temperature is over 25 Â°C, set to the higher value. TEMP 80 FL Display EJECT Test mode Status Remarks
2 3 When 4
The microcomputer starts measuring. End 80 is the default value (unadjusted).
: Measuring value : Set value
Âµ-com Measuring value Room Temperature [ T (Â°C) ] 11.6 to 16.9 17.0 to 22.3 22.4 to 27.6 27.7 to 33.0 33.1 to 38.4 ( hex= â +1 to +3 0 to +2 â1 to +1 0 to â2 â1 to â3 )
Âµ-com Set value
79 When the 7A microcomputer 7B value is 7C 7D
Change with the F. SKIP key (UP) and the R.SKIP key (DOWN). Power OFF
TEMP 7B 7B EEPROMW Writing to EEPROM is made and return is made automatically from test mode to normal mode.
2 Laser Power Check
Step No. 1 Operation Keys and Operation Method Short circuit MECH TEST MODE during the power ON, or press the ESC key. (Service remote control unit)(Fig.1) Press the SET key (Remote control unit) Measure the laser power with an optical power meter. At this time, check that the measuring value (PLAY power) meets the standard value of the table on the right. Note: When the standard value on the right is applied, the numerical value immediately after entry into test mode 2 shall be taken as the standard. FL Display EJECT Test mode Status Remarks Laser Wavelength: 780 nm
2 3 4
ppw ppw ppw
PLAY power output status
Âµ-com Measuring value
May be omitted.
Âµ-com Set value
hex = â +02 to +04 â01 to +01 â02 to â04
Standard Value (mW) 0.69Â±0.12 0.67Â±0.1 0.63Â±0.1
Press the SET or SEARCH key. At this time, check that the measuring value (REC power) meets the standard value of the table on the right.
REC power output status hex = ( â ) +02 to +04 â01 to +01 â02 to â04 The LD goes out. Standard Value (mW) 6.18Â±0.93 5.79Â±0.87 5.47Â±0.82
Press the EJECT key.