There are currently no product reviews.
Some of the pictures in this manual are a bit irritating. I had to dissassemble the unit and some of the screws have different threads, which is not mentioned in this manual. Also some of the drawings of the boards look different than the actual boards.
After all, the manual was very useful. I was able to recalibrate the capstan drive and it is working fine again.
This manual is very good. 303 pages scanned in a very high resolution. My camera has bad, leaking capacitors which all of the V5000 models are suffering from these days.
There is a huge part list with all capacitors, transistors etc. in this manual which helped me a lot. Otherwise I would not have been able to buy replacement parts.
The dissassembly guide is very enormous and detailed. Unlike on the Panasonic MS1 manual I downloaded here it actually looks like the real parts look. And the screws are labeled correctly, so you shouldn't have any left after the repair. ;)
has all the schematics you could need,and very well laid out format also has all part numbers along with an exploded view which is helpful
Very nice to have! Now it is no problem to understand how it is put together.
Helps me a lot.
good scans, all is clear. all pages in order. recommended
6.1.3 MD Mechanism Assy Adjustment and Check
1 Temperature Check (Please perform this check soon after the power has been switched on.)
Note: When IC101 (RF-AMP) or IC104 (mechanical controller) has been exchanged or when a correction changing VD+3 has been made, be sure to correctly perform all steps up to step 7.
Step No. 1 Operation Keys and Operation Method Short circuit MECH TEST MODE during the power ON, or press the ESC key. (Service remote control unit)(Fig.1) Press the REC key. When is 80 or more at this time, the adjustment is OK (step 4 and following can be omitted). is 80 (not passed) Measure the ambient temperature T (Â°C) andcheck hat the measuring value ( ) is within the standard in regard to the microcomputer set value ( ). Note: When the indication of the measuring value ( ) is not stable, and the ambient temperature T (Â°C) is 25 Â°C or less, set to the lower value. When the ambient temperature is over 25 Â°C, set to the higher value. TEMP 80 FL Display EJECT Test mode Status Remarks
2 3 When 4
The microcomputer starts measuring. End 80 is the default value (unadjusted).
: Measuring value : Set value
Âµ-com Measuring value Room Temperature [ T (Â°C) ] 11.6 to 16.9 17.0 to 22.3 22.4 to 27.6 27.7 to 33.0 33.1 to 38.4 ( hex= â +1 to +3 0 to +2 â1 to +1 0 to â2 â1 to â3 )
Âµ-com Set value
79 When the 7A microcomputer 7B value is 7C 7D
Change with the F. SKIP key (UP) and the R.SKIP key (DOWN). Power OFF
TEMP 7B 7B EEPROMW Writing to EEPROM is made and return is made automatically from test mode to normal mode.
2 Laser Power Check
Step No. 1 Operation Keys and Operation Method Short circuit MECH TEST MODE during the power ON, or press the ESC key. (Service remote control unit)(Fig.1) Press the SET key (Remote control unit) Measure the laser power with an optical power meter. At this time, check that the measuring value (PLAY power) meets the standard value of the table on the right. Note: When the standard value on the right is applied, the numerical value immediately after entry into test mode 2 shall be taken as the standard. FL Display EJECT Test mode Status Remarks Laser Wavelength: 780 nm
2 3 4
ppw ppw ppw
PLAY power output status
Âµ-com Measuring value
May be omitted.
Âµ-com Set value
hex = â +02 to +04 â01 to +01 â02 to â04
Standard Value (mW) 0.69Â±0.12 0.67Â±0.1 0.63Â±0.1
Press the SET or SEARCH key. At this time, check that the measuring value (REC power) meets the standard value of the table on the right.
REC power output status hex = ( â ) +02 to +04 â01 to +01 â02 to â04 The LD goes out. Standard Value (mW) 6.18Â±0.93 5.79Â±0.87 5.47Â±0.82
Press the EJECT key.