There are currently no product reviews.
The service manual is scanned, but high quality, high resolution. The smallest texts on schematics diagram is readable. Super! Thank you!
I am satisfied with the service. And if need another manual, i will definitely buy from this site. Keep up the good work.
have download a number of manuals todate , most are excellant, one or two sometimes a little difficult to read but a least avaialable, great site .
Excellent had everything I wanted, very happy with purchase
This service is relatively cheap, document is fast available, schematic is readable.
DV-505, DVL-909, DV-S9
1.2.5 Disc Determination
Determination is achieved by checking the sine wave by sweeping the lens with the OE IC gain at L and the FSC error amplifier (SGC) at the default setting. If no sine wave is detected, checking is retried after switching the OE IC gain to H and increasing the gain of the FSC error amplifier (SGC). If no sine wave is detected again, it is regarded as the NO DISC condition. If one half of the sine wave detected at the first lens sweep is of a value less than 0.5 V, the OE IC gain is set to H and the peak-topeak value of the sine wave is roughly adjusted to 1.8 Vp-p. By sweeping the lens around the height where the sine wave has been detected, disc determination is performed, and the sine wave is finely adjusted to 1.8 Vp-p.
Checking the sine wave None OEIC gain = H FCS AMP gain up
Less than 0.5V
More than 0.5V
OEIC gain = H
Checking the sine wave None NO DISC
Roughly adjust to 1.8Vp-p sine wave
Finely adjust to 1.8Vp-p sine wave END